Definition of training group
The therapist can create a training group of up to five patients per one custo kybe. In addition to patient data, the upper and lower limits of heart rate can be determined for each patient individually. Afterwards the defined training group is imported into custo kybe from the PC via USB.
Beginning of training
All participants of the training group automatically connect themselves to custo kybe and custo guard continuously transmits the ECG. On the display of custo kybe each participant is presented with name and current heart rate. The therapist can start the training by pressing the start key.
Performance of training
After having activated the start key, custo kybe starts recording the ECG signals. In the overview display patient name and the corresponding heart rate are always shown. If heart rate exceeds its upper limit or falls below its lower limit, an acoustic alarm signal will be issued and the patient concerned will be highlighted in red.
ECG display during the training
By clicking on the patient in question, it is possible to change from the overview page to the ECG display of this patient.
The amplitude of the ECG is freely scalable.
With a click on the back button you can switch to the overview page.
Termination of training
In order to finish the training it is sufficient to click on the stop key. The recording is then terminated.
Evaluation of training
The stored ECG data from the training are transferred from custo kybe to the custo Rehab Software via USB or SD card. They can be used to the full extent of the Rehab software. After the data download custo kybe can be programmed for a new training group.